Waveform measurements: Cursors Waveform and Screen. Automatic measurements 29, of which up to four can be displayed on-screen at any one time. Measurements include: Period, Frequency, Delay, Rise Time, Fall Time, Positive Duty Cycle, Negative Duty Cycle, Positive Pulse Width, Negative Pulse Width, Burst Width, Phase, Positive Overshoot, Negative Overshoot, Peak to Peak, Amplitude, High, Low, Max, Min, Mean, Cycle Mean, RMS, Cycle RMS, Positive Pulse Count, Negative Pulse Count, Rising Edge Count, Falling Edge Count, Area and Cycle Area. Gating Isolate the specific occurrence within an acquisition to take measurements on, using either the screen, or waveform cursors. Waveform math Arithmetic Add, subtract, and multiply waveforms. FFT Spectral magnitude. Set FFT Vertical Scale to Linear RMS or dBV RMS, and FFT Window to Rectangular, Hamming, Hanning, or Blackman-Harris.
You may also like
Sign Up for our Test & Measurement Tips Enewsletter: